Fluke Corporation is recalling certain Digital Clamp Meters manufactured between Sept. 1, 2010 and Oct. 31, 2012.
The printed circuit assembly in these units may not be properly fastened to the test lead input jack. This may result in inaccurate voltage readings, including a low or no voltage reading on a circuit energized with a hazardous voltage, presenting a shock, electrocution, or thermal burn hazard.
This recall involves Fluke Digital Clamp Meters with the model and serial numbers listed below:
Starting Serial Number
Ending Serial Number
Sept 2010 – Feb 2011
Sept 2010 – Mar 2011
Oct 2010 – Feb 2011
Oct 2010 – Oct 2012
Any serial number preceded or followed by the letter "R" is NOT affected by this recall notice.
If you own one of these clamp meters, stop using it immediately and send it back to Fluke for repair. (send only the clamp meter; any test leads/boxes/manuals that are sent with the unit will NOT be returned.)
To contact Fluke for information on how to have your units repaired, call 877-564-7180 and request a postage-paid mailing envelope be sent to you.